1. How does the estrous detector work?
The estrous detector picks up electrical resistance in the vaginal mucus, if there is a rise in resistance it indicates ovulation. The detector is inserted into the vagina, testing should be done once or twice a day for several successive days for fail safe ovulation detection.
2. Does the way of using the estrous detector affect the readings?
Yes, it does. The angle at which the probe is introduced into the vagina as well as the probe length influence the readings. For that reason we recommend to perform measurements always in the same or similar manner.
3. Which animals can the Estrous Detector be used for?
The Draminski estrous detectors can be used for the following animals:
These detectors differ mainly with the probe size and location. The user should remember that the test principle is the same in all versions; however, received measurements and interpretations are different for each species and differ for ages of the female.
4. How can I check if my detector operates properly?
1. What is the difference between the Sonofarm Mini and the Animalprofi?
The Sonofarm Mini can only be used on small livestock and is not able to save any images and no measurements can be made. The Animal profi can be used on small and large livestock and you are able to measure foetuses as well as save images to the memory of the machine and transfer to PC.
2. How soon can I start detecting pregnancy with the Sonofarm Mini?
Testing can start as early as 30 days after insemination in sheep. In pigs detection can be done as early as 25 days after insemination.
3. How long does a fully charged battery last on the Sonofarm Mini?
The Sonofarm Mini is supplied with 2 rechargeable batteries. Each battery lasts for 4 hours of continuous work. The batteries are charged for 1 hour and 45 minutes. The Sonofarm Mini will be able to work for 8 hours if both batteries are fully charged.
4. Can the Sonofarm Mini be used on large livestock?
The Sonofarm Mini is only used for small livestock.
1. I would like to consult a given diagnosis with a vet. What can I do?
The Animalprofi Scanner has an internal memory of 200 images. After stopping an image (the START/STOP button) you can save the image in the memory (the Main Menu >Memory>, submenu >Save). In addition, you can describe the image with an animal number and fill in the day of possible pregnancy (date and time will be added automatically). Consequently, you can save important images and consult them, for example, with a vet. Additionally, the scanner is equipped with a possibility to transmit images into a PC. This function allows to save an infinite number of images in your PC and to e-mail or print them, if necessary.
2. Can I use the scanner in small and large animals?
Yes, you can. The Animalprofi Scanner is intended to examine all animals. The user can choose a probe they want the scanner to be equipped with. The abdominal probe is indicated for small animals (pigs, sheep, goats, dogs and cats). The rectal probe is indicated for large animals (cows and mares). The scanner automatically recognizes the type of a probe.
3. Is it possible to measure backfat thickness in pigs?
Yes, it is possible with an abdominal probe. No other probes or additional devices are necessary. When you place the probe on the back of a pig (C7 spot) and locate a correct image of backfat and muscles, freeze the image pressing the STOP/START button. Then open the main menu, select the option >Measure> and then >Length in the submenu. When a red cursor appears, place it in the upper border of the backfat visible in the image and then press the OK button. Then a green cursor appears which should be moved with arrows to the lower border of the backfat. On the right side of the screen a box will appear with the information about the distance between these two cursors (in millimetres) which indicates the backfat thickness.
4. What is the difference between the sector probe and the linear probe?
Both probes are used internally. The linear probe is mostly used for horses as it gives you a clearer image, and a more detailed image. It can also be used on cattle. The sector probe is easier to use as it gives you a wider image of 180° and the linear probe gives you a linear image on the scanner.
1. Is it possible to count the number of somatic cells (somatic cell count) based on the measurement of the electrical resistance of milk?
It is impossible! The instruments measuring how the electricity is conducted in milk and does not count the somatic cells in a direct way. The changes in the resistance level and the number of cells are not directly interconnected. It should be acknowledged that decreasing resistance indicates a developing infection. It is a basic rule which is used to evaluate the health condition of the udder. However, the number of somatic cells reflects the reaction of the immunological system which is specific for each organism. The reaction of this system is a result of the level of stress which a cow is suffering from and good or impaired readiness to fight the infection. Such readiness determines the number of cells in milk and decides when the number of cells reaches a high level after the infection. The measurement of resistance with a manual instrument is a practical method to monitor the health of the udder quarters; however, additional analyses are always necessary in order to make a precise diagnosis and start an efficient treatment in case of disease development.
2. Which stage of udder disease can be checked with the Mastitis Detector?
The most important thing to be aware of is the fact that the DRAMIĆSKI Mastitis Detector measures subclinical conditions of the udder inflammation. It means that with this instrument you can prevent the disease and not allow for clinical development of mastitis. The Mastitis Detector does not detect a clinical form of the disease, so it does not measure the number of somatic cells.
3. What are the main recommendations regarding the use of the Mastitis Detector?
4. Can the Mastitis detector measure tank milk?
No, it can't. Tank milk shows completely different values from the milk from a single cow. Therefore, the interpretation of results presented by the manufacturer can be used only for individual cows; it cannot be used for tank milk.
1. What should I remember while using the Moisture Meter?
2. Can I calibrate my Grain Moisture Meter with my closest Co-op, Supply Depo, Grain Silo or Customer?
Yes you can. Get a sample from the Co-op and get the calibration with their moisture meter. It is possible to then calibrate your Grain Moisture meter to the same as the Co-op.
3. How many different grains can the Grain Moisture Meter test?
The Grain Moisture Meter is able to test 30 different species of grain.